The delivery price is calculated during checkout based on your location and the goods ordered. Our basic delivery prices are:
Delivery within Netherlands, The: €27.00
Delivery Service: Three working days*
* For products that are in stock and the order received by midday.
Some products go through a final configuration and calibration before being shipped. If the calibration labs are busy this can add up to two days on to the delivery schedule but guarantees a fresh calibration.
The CR514 Acoustic Calibrator meets ANSI S1.40:2006 and IEC 60942:2003 to Class 2. It is most suited to the Class 2 sound level meters that NoiseMeters offers.
This calibrator is included in most of the Class 2 Noise Measurement Kits, along with the sound level meter, carrying case and software where applicable. Please refer to the sound level meter web pages for more information.
As soon as the calibrator is switched on it emits a 1 kHz tone at 94 dB. The calibrator should be pushed carefully (firmly, but slowly) on to the microphone capsule.
The meter should now be adjusted accordingly. There will often be a free-field adjustment to make for the microphone in use. So, for example, you may actually be adjusting to 93.8 dB on the meter. For meters provided by NoiseMeters this is either dealt with automatically by the meter, or is covered in the meter's User Manual.
|IEC 60942:2003 Class 2
|94.0 dB ± 0.4 dB
|1 kHz ± 1%
|23°C, 101.3 kPa, 65% RH
|½" (to IEC 1094-4)
|QD7 for use with 1/4" microphone capsules
|1 x 9v (PP3, 6LR61, 6F22)
|121 x Ø47 mm
|220 g with battery
|Push button start, Automatic switch off with green
|LED indication. User override of automatic switch off